Osciloscopio fnirsi 1014d - generador de señal DDS de osciloscopio digital en dos en uno, ancho de banda de 100 MHz de 2 canales 1gsa / s tasa de muestreo
About this deal
High Intelligent Decoding Algorithm: The oscilloscope has an infinite encoder life and intelligent anti-burn function, enabling it to withstand up to 400V in 1X mode.
FNIRSI 1014D - 2 in 1 digital oscilloscope: DDS signal gener FNIRSI 1014D - 2 in 1 digital oscilloscope: DDS signal gener
Complete triggering function (single, normal, automatic), and at any time, the waveform display (STOP function) is frozen. Generous Storage Options: 1GB built-in storage accommodates up to 1000 screenshots and waveform data sets for easy recall. Gear selection: the battery voltage is generally below 40V, other DC voltage is uncertain, so it is necessary to adjust the gear according to the actual situation. If it is higher than 40V (both probe and oscilloscope are set to the same gear)
and in this repository there is a new firmware file for the scope. This is for the scope version with the W25Q32 flash chip (32Mbit). This is also the scope
FNIRSI 1014D - How to upgrade the firmware - TechCorner.TV FNIRSI 1014D - How to upgrade the firmware - TechCorner.TV
Equipped with a USB interface, which can be connected to a computer to share its screenshots with the computer, which is convenient for secondary analysis. On the other hand, everyone’s use cases are different. Besides that, your choice of a tool as integral to your day as a scope or a soldering iron is pretty personal. You like what you like, and it doesn’t always have to make sense.Press the [auto] button once to display the waveform of the crystal vibration tested. If the waveform after automatic adjustment is too small or too large, the waveform size can be adjusted manually by knob PWM signal measurement of MOS tube or IGBT Wrote a lot of the display functions and the touchpanel interface functions. Now I found a lot of the variables used in the scope, Firstly, the oscilloscope is set to normal normal trigger mode (auto trigger mode is default after startup). Normal trigger mode is specially used to measure the non periodic digital signal. If the auto trigger mode is not able to grasp the non periodic signal, the infrared remote control signal belongs to the non periodic digital coding signal